Please Note a valid prescription is required.
The AutoSet series of Automatic Positive Airway Pressure (APAP) machines have long been designed to maximize comfort while providing minimal therapy pressures at the optimal moment. ResMed builds upon the AutoSet Advantage algorithm and incorporates EPR with Easy-Breathe technology in a new post modern shell. The S9 Series aesthetics are a complete redesign with new features.ResMed had incorporated Easy-Breathe Technology along with an Enhanced AutoSet algorithm to make breathing through a CPAP feel more “natural”.
Pressure: 4 to 20 cm H2O (.2 increments)
Ramp Time: 0 to 45 min. (5-min. increments)
Starting Ramp Pressure: 4 to min pressure
Dimensions: 6 in x 5.5 in x 3.25 in
Bag Dimensions – Outside: 13.5 in x 12 in x 5.5 in
Machine Weight: 1.79lbs
Filters: Pollen or Hypoallergenic
Sound Level: 26 dBA
Operating Altitude: Sea level to 8500′ (2591 m)
SlimLine Air Tubing: Flexible plastic, 1.83m, 15mm inner diameter
Optional ClimateLine Tubing Length: 78 inches
Optional ClimateLine MAX Tubing Length: 75 inches
Housing Construction: Flame retardant thermoplastic
Humidification: Optional Integrated Heated Humidifier or Pass-over Humidifier
DC Power: Only use DC Converter by ResMed for DC input or damage may occur.
Warranty: 2 Year Manufacturer
90W Power Supply:
- Dimensions of Power Supply Only – Not including Cord or Plug: 5 1/2 inch (L) x 1 7/8 inch (W) x 1 1/4 inch (D)
- Weight of Power Supply, Cord & Plug: 1.29 Pounds
- Unit Input Range 100-240V, 50-60Hz,
- Nominal for Aircraft Use 110V, 400Hz
- Typical Power Consumption 70W (80VA)
- Maximum Power Consumption 110W (120VA)
European Voltage Usage: Yes, outlet adapters required
Data Storage Capacity:
- Display: 365 days summary data (up to 500 events per session.)
- SD Memory Card: 365 days summary data and 30 Days and 7 nights detailed data
Aircraft Use: These S9 devices comply with US FFC Part 15, Class B requirements if no external data cables are used.
Max Exposure Heat for Machine: Machine should not be exposed to heat greater than 140 degrees Fahrenheit
Required Settings: This machine does not have required settings.
Unless directed otherwise, CPAP.com will leave this machine at its default settings of:
- EPR: Off
- Ramp: On
- Ramp Time: 20 Minutes
- Ramp Start: 4 cm H2O